CE100-80 Verification Masters
- The selected measurement instrument (LMI 200, Digital Indicator, etc..) once turned on is inserted into the 3rd axis probe body and locked in.
- Insert 3rd axis probe into the Verification Master.
- The reading on the measurement instrument should be “.00 +/-.01 mm”
- If readings beyond acceptable limits are observed, simply reset zero on the measurement instrument.
Shipped with CE100-80
NIST traceable certification documents.
Note: The CE100-80 Verification Master checks 3rd Axis Probes with 1.00” dia shafts and 80.0 mm part / surface offsets.