CM1.00 Certification Masters
The CM1.00 Certification Master is used to check and certify the accuracy of the TruPosition Probe. If you have a calibration system and want to maintain files on your TruPosition Probes, then the CM1.00 Certification Master is the best and quickest way to check calibration.
The TruPosition Probe must:
- be set to zero using the Zero Master.
- be inserted into the Certification Master.
- be rotated in the Master until the “A” axis reads “0.00.” the “B” axis should read “.50 +/-.01mm
- be rotated in the Master until the “B” axis reads “0.00.” the “A” axis should read “.50 +/-.01mm
Shipped with CM1.00
NIST traceable certification documents.
Note: Readings are to be taken at all four quadrants( +A, -A, +B, and –B).
The CM1.00 Certification Master checks TruPosition Probes with 1.00” dia shafts.
For CAD files, right click on link select "Save Target As"